LIGO Document D1700326-v1
- Electronics interface to the Thorlabs optical fiber switch. The interface is integrated into an optical table feedthrough panel.
- REDLINE: Found an issue with rev 1 which causes a short between 24V and GND when the switch is engaged. Pins 3 and 4 are swapped on J3. Modify PCB before re-running boards.
DCC Version 3.4.0, contact
Document Database Administrators