LIGO Document G1800345-v1
- The Hanford ITMx was removed after O2 due to a high absorption spot. The optic had been examined and cleaned in-chamber, May 2017. The absorption persisted.
A feature was found at the location predicted by TCS. This feature is highly absorbing. A feature at this location was present in the scatter scans of the optic as received from the coater. Several other post-coating absorbers over 20 ppm were found in the center 60 mm of the optic.
LIGO-E1000766: ITM03 Characterization
LIGO-T1700193: ITM03 inspection at LHO and investigation of circular features
- held from 19 Mar 2018 to 22 Mar 2018 in Sonoma State University
DCC Version 3.4.0, contact
Document Database Administrators